When referencing this material, please cite the following publication:
In addition, please include the standard PhysioNet citation:
The Global Electrical Heterogeneity (GEH) concept is based on the theory of Wilson’s electrical gradient vector, which characterizes the degree of heterogeneity of the total recovery time across the ventricles.
The larger the degree of heterogeneity of total recovery time across the ventricles, the larger the spatial ventricular gradient (SVG) magnitude. The SVG vector points towards the area where the total recovery time is shortest. SVG vector points the direction along which non-uniformities in excitation and repolarization are the greatest (i.e., it is perpendicular to the line of conduction block). Experimental and theoretical investigations demonstrated that the SVG is related to global heterogeneity of both action potential duration and morphology.
The concept underlying the SVG was extended to the spatial QRS-T angle, the three-dimensional angle between the QRS- and T-vectors and the sum absolute QRST integral (SAI QRST), a scalar analog of the SVG calculated as the absolute value of the area under the QRS complex and T-wave on the X, Y, and Z leads. The scalar value of SVG can also be calculated as a QT integral on Vector Magnitude signal (iVMQT), as an area under the Vector Magnitude signal curve from the QRS-onset to T-offset. Five GEH metrics (SVG magnitude, elevation, and azimuth, spatial QRS-T angle, and SAI QRST (or QT integral on Vector Magnitude signal, iVMQT) are complementary to each other; all together they characterize global electrophysiological properties of the heart. GEH is independently associated with sudden cardiac death. GEH can be measured on routinely used clinical 12-lead ECG, after its transformation into orthogonal (Frank) XYZ ECG. We recommend using Kors transformation.
Software Description and Usage
This page contains V.1 of the software. The working repository for this is hosted in the following github page: https://github.com/Tereshchenkolab/Global-Electrical-Heterogeneity.
One test file
90757.mat is provided for GEH calculation testing, with a sampling rate 500 Hz and amplitude resolution 1 µV. A raw 12-lead ECG file
12LECG.mat is provided to illustrate the Kors transformation from 12-lead to XYZ (Frank) ECG.
For a demonstration, load
12LECG.mat and run
Kors_git.m, then load
90757.mat and run
Erick Andres Perez Alday, PhD, email@example.com Annabel Li-Pershing, BS, firstname.lastname@example.org Muammar Kabir, PhD, email@example.com Larisa Tereshchenko, MD, PhD, firstname.lastname@example.org
Name Last modified Size Description
Parent Directory - LICENSE 2018-03-30 10:34 1.0K Kors_git.m 2018-03-30 10:34 2.3K 90757.mat 2018-03-30 10:34 12K GEH_analysis_git.m 2018-03-30 10:34 23K 12LECG.mat 2018-03-30 10:34 109K GEH calculation.pdf 2018-03-30 10:34 267K
If you would like help understanding, using, or downloading content, please see our Frequently Asked Questions.
If you have any comments, feedback, or particular questions regarding this page, please send them to the webmaster.
Comments and issues can also be raised on PhysioNet's GitHub page.
Updated Friday, 28 October 2016 at 16:58 EDT